(1)Choose an optimal position and fix the ion bar and power supply tightly.
(2)Insert the HV plug on the bar body into the HV output connection on power supply.
(3)Connect the ground end on the bar body to the ground bolt on the power.
(4)Switch on the power.
2.Outline dimensional drawing
User should clean and maintain timely to ensure the good performance of the ion bar according to working environment and requirement of ESD protection. Clean the carbon materials on election bare tip, discharge pad and metal discharge body with static brusher or clean cloth with ethanol can obviously improve the performance.
Must be done after the power is shut done for 10 minutes
Restart must wait till alcohol completely volatilized after cleaning by clean cloth with ethanol. Any other organic solution is not allowed to be used to clean the ion bar.
Stop using the device when red indicator turns off on power supply. Re-use the device only after checking and repairing by professionals.
After Sales Service
(1)AP-AB1103 ion bar is strictly inspected before ex-work and match the entire related technical standard indicated in this manual.
(2)Guarantee：Repair or change any damaged parts caused by quality reasons free of charge within one year from the date of purchase after we confirm the quality issue.
(3)Above guarantee is not fit for the following situation：Alternation of the device, wrongly used, incorrect installation, abuse, negligence, and accidental damage, wrong input voltage, unpacking and repacking by user itself or maintain by other unauthorized service department.
(4)AP&T is free from any related responsibility of using the product except the parts specified here above.